Gives electroacoustical performance specifications for three kinds of sound measuring instruments: - a conventional sound level meter that measures exponential time-weighted sound level; - an integrating-averaging sound level meter that measures time-average sound level; and - an integrating sound level meter that measures sound exposure level. A single instrument may make any, or all, of the three kinds of measurements. Two performance categories, class 1 and class 2, are specified in this standard. In general, specifications for class 1 and class 2 sound level meters have the same design goals and differ mainly in the tolerance limits and the range of operational temperatures. Tolerance limits for class 2 specifications are greater than, or equal to, those for class 1 specifications. Sound level meters conforming to the requirements of this standard have a specified frequency response for sound incident on the microphone from one principal direction in an acoustic free field or from random directions.
Directives related to this standard.
Directive 2004/108/EC of the European Parliament and of the Council of 15 December 2004 on the approximation of the laws of the Member States relating to electromagnetic compatibility and repealing Directive 89/336/EEC
Directive 2014/30/EU of the European Parliament and of the Council of 26 February 2014 on the harmonisation of the laws of the Member States relating to electromagnetic compatibility (recast)